Used primarily to measure Undercut, but can also be used to measure porosity. Also suitable as a depth gage.
- Depth Gage
- Measures Porosity & Undercut
Gages come with certified calibration documentation highlighting calibration was performed in accordance with ISO 17025:2017, ANSI/NCSL Z540-1, Mil-STD- 45662A, Webber Quality Manual QCM-1G dated 11/15/14, and Webber Calibration Procedure WCP-C13 9/6.2006. Traceable to NIST. The numbered and registered gages are manufactured and tested to tolerances not to exceed +/- .006 (or +/- .15mm) tolerance level. These gages meet or exceed all requirements for the American Association of Railroads and the Federal Railroad Administration. These gages are direct contact instruments and some wear will occur after usage. The purchaser should periodically re-calibrate these gages or return them to Atema for recalibration and recertification. This assures that excess wear has not compromised the accuracy of the instruments.
A note on calibration dates:
Calibration dates are suggested by the manufacturer. You may overwrite the one year re-calibration date with an interval of your choosing, as long as this is documented in your Quality Manual or other quality documents. Atema recommends a 5 year re-calibration date for gages used as a Master set.